- Open Access
Comparison of a flexible versus rigid bone cement injection system in unilateral percutaneous vertebroplasty
European Journal of Medical Research volume 25, Article number: 36 (2020)
Percutaneous vertebroplasty (PVP) has been demonstrated to be effective in the treatment of acute osteoporotic vertebral fracture (AOVF). However, bilateral puncture takes more time to accept more X-ray irradiation; some scholars apply unilateral puncture PVP, but the cement cannot be symmetrically distributed in the vertebral body, so we use a flexible cement injector that undergoes PVP through the unilateral pedicle puncture. This research aims to compare the clinical results of PVP for AOVF with unilateral pedicle puncture using a straight bone cement injector and a bendable cement injector, determine the value of a bendable cement injector.
We undertook a retrospective analysis of patients with thoracic and lumbar compression fracture treated with unilateral pedicle puncture percutaneous vertebroplasty from our institution from June 2013 to July 2015. Operation time, radiation exposure, bone cement injection amount, and the incidence of bone cement leakage were recorded on presentation, the cement leakage was measured by X-ray and computed tomography scan. The patients were followed up postoperatively and were assessed mainly with regard to clinical and radiological outcomes.
There was no significant difference in the operation time, radiation exposure time and incidence of bone cement leakage between the two groups. There was significant difference in the amount of bone cement injection and the difference between the two groups. There were no significant differences in VAS and the relative height of the vertebral body and local Cobb angle and QUALEFFO between the two groups at 1 week after PVP, significant difference was observed only 12 months after operation.
Application of flexible cement injector is safe and feasible, compared with the application of straight bone cement injector, without prolonging the operative time, radiation exposure time and the incidence of bone cement leakage; it has the advantages of good long-term effect and low incidence of vertebral fracture recurrence.
Osteoporotic vertebral fracture is one of the most common diseases of the elderly [1,2,3,4]. Percutaneous vertebroplasty (PVP) is widely used to treat painful vertebral compression fractures and strengthen the stability of vertebrae [5, 6]. The usual standard in introduction of bone cement in PVP requires a bilateral pedicle approach to create a symmetrical distribution of bone cement . However, pedicle puncture from both sides almost doubles the operation time and radiation exposure compared with a unilateral approach. Reducing operative time and radiation exposure is a valid objective. However, there is controversy about the efficacy of a unilateral approach. One study has reported similar short-term efficacy to bilateral procedures . Another has suggested that introduction of bone cement unilaterally may lead to asymmetric loading of the vertebral body and collapse of the contralateral side of the vertebral body under axial compression stress .
Materials and methods
The study protocol was approved by the Institutional Review Board and the Ethics Committee of Luohe Medical College.
We undertook a retrospective analysis of patients with thoracic and lumbar compression fracture treated with unilateral pedicle puncture PVP from our institution over a 2-month period (June and July 2015). 78 patients were included according to the standard. Inclusion criteria includes: ① age from 60 to 99 years; ② bone attenuation (T score < − 2.5) on bone densitometry; ③ collapse more than 15% of the vertebral height; ④ severe back pain related to a single-level AOVF refractory to analgesic medication; ⑤ using magnetic resonance (MR) imaging, the affected vertebral body showed a hypointense signal on T1-weighted images and hyperintense signal on T2-weighted images. The exclusion criteria included: ① secondary osteoporosis (corticosteroids, endocrine disorders and an inflammatory process); ② failure to acquire informed consent; ③ uncorrected coagulopathy; ④ systemic or local spine infection; ⑤ painless AOVF; ⑥ spinal metastatic cancer; ⑦ severe comorbidities of the cardiorespiratory, hepatic, renal or neurological symptoms. Patients were divided into two groups according to surgical procedure: a flexible cement injection group (36 cases) using a flexible-tipped bone cement injection and 3-point cement injection technique; a rigid bone cement injection group (42 cases) using a straight bone cement injection technique.
Flexible bone cement injection equipment (Ningbo Branch Huakerun Biotechnology Co., Ltd) with angled bone cement injector. Rigid bone cement injection equipment (Shandong Guanlong Medical Products Co., Ltd).
All the PVP procedures were performed in the operating theatre. Patients were placed prone, supported by two transverse bolsters under thorax and pelvis. Gentle distraction and extension was applied to reduce the vertebral fracture. During the procedure, a unilateral transverse process–pedicle approach was adopted with application of local anesthesia. The entry point in the vertebra was identified by fluoroscopy at the junction of the lateral edge of the pedicles and vertebral plate. The trocar penetrated cortical bone at the lateral edge margin of the vertebral arch, and was advanced medially and inferiorly. Fluoroscopy was used to confirm that the needle tip reached the posterior wall of the vertebral body. No further advance was made beyond about 4 mm anterior to the posterior surface of the vertebral body. During the procedure all patients were observed closely with frequent fluoroscopy and the cement injection was stopped immediately if bone cement leakage occured.
When using the rigid cement injector, an 11–13G core needle was advanced from a posterolateral entry point through the involved vertebral pedicle to the junction of anterior and middle thirds of the vertebral body. The inner core was retracted and 3–4 ml PMMA was injected guided by continuous fluoroscopy. When bone cement began to fill the posterior third of the vertebral body, the injection was terminated. In contrast, the flexible bone cement injection method is more involved; detailed descriptions of the process, can be found in the legends of Figs. 1 and 2.
After injection of bone cement, all injection components were withdrawn and pressure was applied to the wound for haemostasis. All the patients were observes supine for 6 h. The next-day rehabilitation included sitting and standing as tolerated. Bisphosphonates were generally used to treat osteoporosis after surgery.
The operation time, radiation exposure time, the amount of bone cement injection and the leakage of bone cement were recorded for each patient in two groups. Clinical assessments were evaluated before surgery, 1 week after surgery and 12 months after surgery. Radiographs and computed tomography (CT) scans were performed to assess the cement leakage in the vertebral body and other possible local complications, and all the complications and adverse events were recorded.
On pain measurement, VAS scores were used which ranged from 0 (no pain) to 10 (worst pain ever). Quality-adjusted life years (QALYs) and the Quality of Life Questionnaire of the European Foundation for Osteoporosis (QUALEFFO) were investigated in all patients, which comprise a 41-item questionnaire organized into 5 domains (Pain, Physical Function, Social Function, General Health Perception, and Mental Function). Each domain’s score and QUALEFFO total scores are recorded on a 100-point scale, lower scores corresponding to better health-related quality of life.
Anteroposterior and lateral standing radiographs were observed to measure vertebral height and kyphotic angle of the vertebral body of all patients in three periods (preoperatively, 1 week after surgery, and 12 months after surgery). In the X-ray radiographs, the anterior height of the affected vertebral body and adjacent normal vertebrae were measured, and the relative anterior height (RAH) of the fractured vertebra was calculated according to the equation:
RAH = fractured vertebral anterior height/[(superior vertebral anterior height + inferior vertebral anterior height)/2] × 100%.
The kyphotic angle was based on the Phillips method, the angle between the superior endplate at one level above the fractured vertebrae and inferior endplate at one level below the fractured vertebrae were measured (Fig. 3).
All statistical analyses were performed with the use of SPSS software, version 12 (SPSS Inc., Chicago, IL). The results were expressed as average ± SD. One-way analysis of variance (ANOVA) was used to compare the VAS scores, quality of life, RAH, and the kyphotic angle between the 2 groups. Difference in cement leakage rate of 2 groups was assessed using χ2 test. P < 0.05 was considered to have statistical significance.
All surgeries in two groups were completed successfully, and no intraoperative deaths were reported in this study. The average duration of follow-up was 15.8 months (range from 12 to 32 months). In terms of demographic data of patients, no significant difference was found between the two groups (Table 1). Typical cases are shown in Figs. 4, 5.
There was no significant difference between two groups in the operation time and radiation exposure time (P > 0.05). The operation time was 28.4 ± 2.82 min (min) and 26.6 ± 2.35 min in the flexible cement injection (subject) group and rigid cement injection (control) group, respectively. In the flexible cement injection group, the radiation exposure time was 4.71 ± 0.95 min, compared with the radiation exposure time of 4.68 ± 0.80 min in the rigid cement injection group.
The average volume of the injected cement was 5.5 ± 0.35 ml and 3.2 ± 0.38 ml in flexible cement injection group and rigid bone cement injection group, respectively. A statistically significant difference was found between the two groups (P < 0.05) (Table 2).
No statistically significant differences were found in the baseline VAS and quality of life scores in the two groups. All scores were reduced in both groups after PVP surgeries, and there was no statistically significant difference between two groups at 1 week after PVP. However, statistically significant differences were found at 12 months after surgery (P < 0.05) (Table 3).
Preoperative and postoperative radiographical assessments of two groups were measured, and documented in Table 4. The RAH in the treatment group increased from 41.06 ± 5.58% preoperatively to 44.30 ± 3.80% at 1 week post-operation, and was 43.04 ± 5.19% 1 year postoperatively. In the control group, RAH increased from 43.01 ± 5.42% preoperatively to 45.49 ± 4.25% at 1 week post-operation, but was 30.86 ± 4.55% at 1 year post-surgery. There was no significant difference in RAH either preoperatively or 1 week postoperatively (P > 0.05). However, there were statistically significant differences between two groups after 12 months.
The Cobb angle in the treatment group changed from 24.15 ± 4.07° before surgery to 22.68 ± 3.26° at 1 week after surgery and 23.14 ± 3.53 at 1 year. In the control group it decreased from 23.66 ± 3.35° to 22.60 ± 3.38° at 1 week after operation, but had increased to 31.36 ± 5.41° at 1 year. There was no significant difference (P > 0.05) between two groups preoperatively and 1 week after surgery, however, a statistically significant difference was found between two groups after 12 months of follow-up, with the Cobb angle of the treatment group being smaller than that of the control group (Table 4).
No procedure-related adverse events were observed in this study. Some extra-vertebral cement leakages were found in the intraoperative and postoperative radiographs. According to radiographic analysis, the rate of cement leakages was 13.9% (5 of 36) and 14.3% (6 of 42) in the therapy group and control group, respectively, and no significant difference was found between two groups (P > 0.05).
Feasibility, safety and clinical efficacy of PVP with flexible cement injectors
Some studies showed that the relief of pain is associated with symmetrical distribution of bone cement in the vertebral body. The improvements of physical status largely depend on the volume of cement infusion during PVP [9, 10]; insufficient cement volume may lead to poor efficacy and failure of surgery. Conversely, increasing the amount of cement injection may increase the incidence of cement leakage [11,12,13]. Clinical studies have shown that 70% of complications of vertebral body forming are associated with the cement leakage [14,15,16,17]. To overcome the shortcomings of PVP technology in the area of sufficient and symmetrical cement distribution, the flexible bone cement injector was designed to deliver bone cement at three zones in the vertebral body. Due to the multi-point injection, the injected cement is in the state of low-pressure diffusion and subsequently bone cement leakage caused by high pressure has been avoided.
In traditional rigid injection cannulae, in order to distribute bone cement evenly on both sides of the vertebral body via the unilateral puncture, the angle of trocar approach should be as oblique as possible without damaging the medial cortex of the pedicle. This may risk neurological damage. This more oblique angle is not necessary in bilateral puncture, but the frequency of operative time and radiation exposure will be increased. These problems are overcome by using the flexible cement system .
In this study, the results confirmed the efficacy of PVP in the treatment of AOVF, which were reflected in immediate and significant change in VAS and QUALEFFO scores. Meanwhile, using flexible bone cement injector and three-region injection technology, it was shown that bone cement can be distributed more evenly in the vertebral body even punctured through single side of the vertebral pedicle.
The influence of asymmetric parameters of vertebral body on PVP
We have not conducted in-depth study on the influence of preoperative vertebral asymmetric parameters on bone cement injection. The effect of PVP is related to the distribution of bone cement in the vertebral body. The ideal bone cement dispersion should be distributed on both sides of the midline of the vertebral body, with symmetrical distribution. If the asymmetric distribution of bone cement increases the transmission of unilateral load, the ideal effect cannot be achieved. There are many factors affecting the distribution of bone cement in vertebral body, including viscosity of bone cement, injection volume, puncture angle, vertebral body-related factors (bone density, vertebral structure, etc.)
Long-term efficacy of the two systems
Murphy  reported that uneven distribution of the bone cement in the vertebral body may lead to further pressure-loading in the injury side of vertebra, and subsequent instability of spine. Under the constant loading, the vertebral body may buckle to the contralateral side, and thus causing further compression deformation of the vertebral body. The flexible bone cement injector cannula bends at its immediate exit from the introducer due to the elastic energy of the inner wire of Ni–Ti alloy, the curved angle channel reaches zones which a traditional rigid cannula cannot, thereby permitting a more uniform strengthening effect of the whole vertebra. In this study, compared with the control group, both vertebral height and back pain were significantly worse in the control group than the treatment group at 1-year follow-up. It seems likely that more uniform cement distribution and increased volume of cement improve the stability of the spine in the long-term.
There were some limitations in this retrospective study, and the sample of patients included is rather small. Moreover, follow-up periods in two groups were relatively short. Long-term follow-up data with a larger sample of patients are needed in future studies.
This study confirmed that both flexible injector and straight bone injector PVP are safe and effective in the treatment of painful AOVF. During the period of follow-up (12 months), both methods showed good clinical outcomes. However, bone cement was distributed more evenly in the vertebral body via the flexible system without prolonging the surgical time, radiation exposure or increasing the incidence of bone cement leakage. The flexible cement injection technology demonstrated better long-term efficacy, compared with the traditional straight bone cement injection technology.
Availability of data and materials
All data generated or analysed during this study are included in this published article.
Acute osteoporotic vertebral fracture
Quality of life questionnaire of the European Osteoporosis Foundation
Relative anterior height of the fractured vertebra
Kallmes DF, Comstock BA, Heagerty PJ, Turner JA, Wilson DJ, Diamond TH, Edwards R, Gray LA, Stout L, Owen S, Hollingworth W, Ghdoke B, Annesley-Williams DJ, Ralston SH, Jarvik JG. A randomized trial of vertebroplasty for osteoporotic spinal fractures. N Engl J Med. 2009;361(6):569–79. https://doi.org/10.1056/NEJMoa0900563.
Buchbinder R, Osborne RH, Ebeling PR, Wark JD, Mitchell P, Wriedt C, Graves S, Staples MP, Murphy B. A randomized trial of vertebroplasty for painful osteoporotic vertebral fractures. N Engl J Med. 2009;361(6):557–68. https://doi.org/10.1056/NEJMoa0900429.
Johnell O, Kanis J. An estimate of the worldwide prevalence and disability associated with osteoporotic fractures. Osteoporos Int. 2006;17(12):1726–33. https://doi.org/10.1007/s00198-006-0172-4(Epub 2006 Sep 16).
Rousing R, Hansen KL, Andersen MO, Jespersen SM, Thomsen K, Lauritsen JM. Twelve-months follow-up in forty-nine patients with acute/semi acute osteoporotic vertebral fractures treated conservatively or with percutaneous vertebroplasty: a clinical randomized study. Spine. 2010;35(5):478–82. https://doi.org/10.1097/brs.0b013e3181b71bd1.
Karmakar A, Acharya S, Biswas D, Sau A. Evaluation of percutaneous vertebroplasty for management of symptomatic osteoporotic compression fracture. J Clin Diagn Res. 2017;11(8):RC07–10. https://doi.org/10.7860/jcdr/2017/25886.10461.
Takura T, Yoshimatsu M, Sugimori H, Takizawa K, Furumatsu Y, Ikeda H, Kato H, Ogawa Y, Hamaguchi S, Fujikawa A, Satoh T, Nakajima Y. Cost-effectiveness analysis of percutaneous vertebroplasty for osteoporotic compression fractures. Clin Spine Surg. 2017;30(3):E205–10. https://doi.org/10.1097/BSD.0b013e3182aa4c29.
Zhang LG, Gu X, Zhang HL, Zhang QG, Cai XB, Tao K. Unilateral or bilateral percutaneous vertebroplasty for acute osteoporotic vertebral fracture: a prospective study. J Spinal Disord Tech. 2015;28(2):E85–8. https://doi.org/10.1097/bsd.0000000000000147.
Cheng X, Long HQ, Xu JH, Huang YL, Li FB. Comparison of unilateral versus bilateral percutaneous kyphoplasty for the treatment of patients with osteoporosis vertebral compression fracture (OVCF): a systematic review and meta-analysis. Eur Spine J. 2016;25(11):3439. https://doi.org/10.1007/s00586-016-4395-6.
Liebschner MA, Rosenberg WS, Keaveny TM. Effects of bone cement volume and distribution on vertebral stiffness after vertebroplasty. Spine. 2001;26(14):1547–54.
Sun H, Li C. Comparison of unilateral and bilateral percutaneous vertebroplasty for osteoporotic vertebral compression fractures: a systematic review and meta-analysis. J Orthop Surg Res. 2016;11(1):156. https://doi.org/10.1186/s13018-016-0479-6.
Kinzl M, Benneker LM, Boger A, Zysset PK, Pahr DH. The effect of standard and low-modulus cement augmentation on the stiffness, strength, and endplate pressure distribution in vertebroplasty. Eur Spine J. 2012;21(5):920–9. https://doi.org/10.1007/s00586-011-2119-5(Epub 2011 Dec 15).
Wang C, Fan S, Liu J, Suyou L, Shan Z, Zhao F. Basivertebral foramen could be connected with intravertebral cleft: a potential risk factor of cement leakage in percutaneous kyphoplasty. Spine J. 2014;14(8):1551–8. https://doi.org/10.1016/j.spinee.2013.09.025(Epub 2013 Oct 17).
Venmans A, Klazen CA, Lohle PN, van Rooij WJ, Verhaar HJ, de Vries J, Mali WP. Percutaneous vertebroplasty and pulmonary cement embolism: results from VERTOS II. AJNR Am J Neuroradiol. 2010;31(8):1451–3. https://doi.org/10.3174/ajnr.a2127.
Kim SM, Kim JU, Lim JH, Jang JS, Jang IT. Delayed complication of vertebroplasty: intradural penetration of previous PMMA leakage after trauma. Spine J. 2015;15(5):1146–9. https://doi.org/10.1016/j.spinee.2015.01.025(Epub 2015 Jan 28).
Corcos G, Dbjay J, Mastier C, Leon S, Auperin A, De Baere T, Deschamps F. Cement leakage in percutaneous vertebroplasty for spinal metastases: a retrospective evaluation of incidence and risk factors. Spine. 2014;39(5):E332–8. https://doi.org/10.1097/brs.0000000000000134.
Hatzantonis C, Czyz M, Pyzik R, Boszczyk BM. Intracardiac bone cement embolism as a complication of vertebroplasty: management strategy. Eur Spine J. 2017;26(12):3199–205. https://doi.org/10.1007/s00586-016-4695-x(Epub 2016 Aug 17).
Hoppe S, Wangler S, Aghayev E, Gantenbein B, Boger A, Benneker LM. Reduction of cement leakage by sequential PMMA application in a vertebroplasty model. Eur Spine J. 2016;25(11):3450–5 (Epub 2015 Apr 5).
Srinivasan D, Than KD, Wang AC, La Marca F, Wang PI, Schermerhorn TC, Park P. Radiation safety and spine surgery: systematic review of exposure limits and methods to minimize radiation exposure. World Neurosurg. 2014;82(6):1337–43. https://doi.org/10.1016/j.wneu.2014.07.041(Epub 2014 Aug 1).
Murphy KJ, Lin DD, Khan AA, Gailloud P. Multilevel vertebroplasty via a single pedicular approach using a curved 13-gauge needle: technical note. Can Assoc Radiol J. 2002;53(5):293–5.
First of all, I would like to extend my sincere gratitude to Professor Wang Haijiao, for his instructive advice and useful suggestions on my thesis. I am deeply grateful to his help in the completion of this thesis. High tribute shall be paid to Mr. Wang Haoran, whose profound knowledge of English triggers my love for this beautiful language and whose earnest attitude tells me how to learn English. I am also deeply indebted to all the other tutors and teachers in Translation Studies for their direct and indirect help to me. Special thanks should go to my friends who have put considerable time and effort into their comments on the draft. Finally, I am indebted to my parents for their continuous support and encouragement.
This work was financed by grants from National Health and Family Planning Commission of the People’s Republic of China (W2014ZT096) and Henan provincial health and family planning commission (201504092).
Ethics approval and consent to participate
The study protocol was approved by the Institutional Review Boards and the Ethics Committees of Luohe Medical College. Before operation, informed consents were signed by all the patients after detailed explanation of the therapeutic procedure to the patients. The study is conducted according to the guideline for case series.
Consent for publication
Written informed consent for publication was obtained from all participants.
The authors declare that they have no competing interests
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Rights and permissions
Open Access This article is licensed under a Creative Commons Attribution 4.0 International License, which permits use, sharing, adaptation, distribution and reproduction in any medium or format, as long as you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons licence, and indicate if changes were made. The images or other third party material in this article are included in the article's Creative Commons licence, unless indicated otherwise in a credit line to the material. If material is not included in the article's Creative Commons licence and your intended use is not permitted by statutory regulation or exceeds the permitted use, you will need to obtain permission directly from the copyright holder. To view a copy of this licence, visit http://creativecommons.org/licenses/by/4.0/. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated in a credit line to the data.
About this article
Cite this article
Li, Y., Cui, W., Zhou, P. et al. Comparison of a flexible versus rigid bone cement injection system in unilateral percutaneous vertebroplasty. Eur J Med Res 25, 36 (2020). https://doi.org/10.1186/s40001-020-00436-z
- Percutaneous vertebroplasty
- Osteoporotic vertebral fracture
- Unilateral puncture