- Open Access
Investigation of the flexibility of bronchoscopes
© I. Holzapfel Publishers 2011
- Received: 16 March 2011
- Accepted: 24 May 2011
- Published: 12 September 2011
For transbronchial biopsy of peripheral pulmonary lesions manouevering the biopsy instrument into acutely angulated subsegmental ostia is frequently problematic. The aim of the present study was to compare the stiffness of various biopsy instruments with regard to their use in the clinical setting.
The maximal anterograde and retrograde bending angles, were measured for various bronchoscopes and biopsy instruments. Measurement was made with the distal tip of the biopsy instrument either flush with the distal end of the bronchoscope, or extending 1.5 cm beyond it. The following scopes and biopsy instruments were investigated: 1. the 6.2 mm outside diameter (OD) bronchoscope, with the 2.4 mm OD forceps, 0.7 mm needle, 2.3 mm catheter, and the 1.9 mm cryoprobe. 2. the 5.1 mm OD bronchoscope, with the 1.8 mm forceps, 0.7 mm needle, and the 1.8 mm catheter. 3. the 3.7 mm bronchoscope, with the 1.0 mm forceps and the 0.8 mm forceps.
Maximum angulation was greater by 35.4° with the needle extended, vis-à-vis the needle "flush". Both needle and catheter were associated with a greater angulation of up to 20.5° in comparison with the forceps. With an instrument in the working channel the largest anterograde angles were measured for the 5.1 mm bronchoscope, and the largest retrograde angles for the 6.2 mm bronchoscope.
When selecting the optimal instrument for transbronchial biopsy specimen collection, account must be taken of the fact that the degree of angulation will depend on the type and diameter of the instrument employed.
- Transbronchial biopsy
- lung cancer
- angulation of biopsy instruments
- needle aspiration
- catheter aspiration
For the diagnostic workup of peripheral carcinoma of the lung employing flexible bronchoscopy, various instruments are available for transbronchial specimen collection [1, 2]. To obtain histological material, transbronchial forceps biopsy (TBB) is employed. Although it is known that the latest technique of cryobiopsy can provide large histological specimens of good quality, no relevant data are currently available for peripheral pulmonary lesions [3, 4]. Both transbronchial needle aspiration (TBNA) and transbronchial catheter aspiration (TBCA) are cytological techniques which with regard to malignant lesions have a "hit rate" higher by at least 15% vis-à-vis forceps biopsy [5–7].
Needle aspiration has the advantage that it can also be applied to extrabronchial processes, such as, for example, haematogenic metastases . In contrast to forceps biopsy, TBCA enables material to be collected not merely directly from the lesion itself, but also from its immediate neighbourhood . Apart from the specific material collection properties of the biopsy instrument, its stiffness, which modifies the bendability of the bronchoscope, is a further major determinative factor with an influence on diagnostic sensitivity. Owing to the pressure-volume relationship in the lungs during inspiration, most carcinomas are localised in the upper lobes. The sharply acute angles met with in the apical upper lobe segments, however, often make it impossible to deploy the biopsy instruments - a difficulty also encountered in the apical lower lobe segments. For the same reason also, biopsy collection from peripheral lesions under endobronchial ultrasound guidance in the apical upper lobe segments has a lower hit rate than for other locations - with the added risk of damage to the US probe [10, 11].
With the aim of facilitating the choice of the most suitable biopsy instrument and bronchoscope diameter for the specific clinical task in hand, the present experimental bronchoscope study compared the maximum angulation of various combinations of scopes and instruments.
We performed our measurements on 3 video bronchoscopes (Pentax Europe, Hamburg, Deutschland), each with a different outside diameter. The angles were measured first without, and then with, the respective biopsy instrument in the working channel. The distal tip of the biopsy instrument was either flush with the distal end of the bronchoscope ("flush"), or projected 1.5 cm beyond it (1.5 cm). The 1.5 cm was determined the length of the needle employed (1.3 cm).
Video endoscope (VE) outside diameter (OD) 6.2 mm (working channel diameter 2.8 mm); instruments: biopsy forceps OD 2.4 mm (oval; Pentax Europe, Hamburg, Germany); cryoprobe OD 1.9 mm (Erbe, Tübingen, Germany); transbronchial biopsy needle OD 0.7 mm (OD needle sheath 1.8 mm, needle length 13 mm; Endo-Technik, Solingen, Germany); Aspiration catheter OD 2.3 mm (Wieser, Egenhofen, Germany).
Video endoscope (VE) outside diameter (OD) 5.1 mm (working channel diameter 2.0 mm); instruments: disposable biopsy forceps OD 1.8 mm (oval; Medwork, Höchstadt/Aisch, Germany); transbronchial biopsy needle OD 0.7 mm (OD needle catheter 1.8 mm, needle length 13 mm; Endo-Technik, Solingen, Germany); aspiration catheter OD 1.8 mm (Wieser, Egenhofen, Germany).
Video endoscope (VE) outside diameter (OD) 3.7 mm (working channel diameter 1.2 mm); instrument: biopsy forceps OD 1.0 mm (oval; MTW, Wesel, Germany); biopsy forceps OD 0.8 mm (oval; Endo-Passion, Dürbheim, Germany).
Each of the bronchoscopes used in the experiment had previously been employed in approximately 500 clinical examinations.
Maximum angulation of biopsy instruments
Maximum Angulation [°]
Anterograde Instruments flush
Anterograde Instruments 1.5 cm
Retrograde Instruments flush
Retrograde Instruments 1.5 cm
Bronchoscope alone OD 6.2 mm
156.9 ± 3.1
110.7 ± 4.3
with forceps OD 2.4 mm
129.1 ± 2.8
132.3 ± 3,3
96.9 ± 2.7
95.7 ± 1.5
with needle OD 0.7 mm
137.0 ± 3.0
148.4 ± 3.1
103.9 ± 3.1
108.3 ± 1.7
with catheter OD 2.3 mm
133.3 ± 1.7
149.9 ± 2.0
101.9 ± 2.4
110.1 ± 2.3
with cryoprobe OD 1.9 mm
119.5 ± 3.5
121.9 ± 1.6
82.5 ± 2.7
88.6 ± 1.1
Bronchoscope alone OD 5.1 mm
203.0 ± 2.0
113.6 ± 1.7
with forceps OD 1.8 mm
176.6 ± 3.6
167.0 ± 3.3
92.9 ± 2.2
90.7 ± 2.9
with needle OD 0.7 mm
145.1 ± 3.6
180.5 ± 2.6
86.3 ± 2.1
102.3 ± 2.0
with catheter OD 1.8 mm
190.1 ± 3.0
187.5 ± 3.8
99.9 ± 2.3
100.8 ± 1.4
Bronchoscope alone OD 3.7 mm
203.9 ± 2.1
105.3 ± 1.1
with forceps OD 0.8 mm
162.0 ± 3.7
157.3 ± 2.4
84.6 ± 2.0
90.6 ± 1.4
with forceps OD 1.0 mm
143.2 ± 6.1
141.4 ± 3.1
64.3 ± 2.3
64.3 ± 1.0
Bronchoscope without biopsy instrument
For anterograde angulation measurement with no instrument in the channel, angles were smaller by between 6.1° (VE OD 3.7 mm) and 23.1° (VE OD 6.2 mm), and for retrograde angulation smaller by between 14.7° (VE OD 3.7 mm) and 19.3° (VE OD 6.2 mm) - vis-a-vis the manufacturer's specification for maximum angulation.
Biopsy instruments "flush" and extended (1.5 cm)
On extending the biopsy instrument, an additional increase in the anterograde angle of 11.4° (VE OD 6.2 mm) and 35.4° (VE OD 5.1 mm) was obtained for the needle; while for the catheter the increase was anterograde 16.6° and retrograde 8.2° (VE OD 6.2 mm).
Comparison of biopsy instruments (VE OD 6.2 mm)
In the extended option, all instruments were associated with larger angles - both up and down. For needle and catheter in comparison with forceps and cryoprobe the respective measurements were anterograde 148.4° / 149.9° versus 132.3° / 121.9° and retrograde 108.3° / 110.1° versus 95.7° / 88.6° - that is, an appreciably greater angulation. With needle or catheter in the channel the difference vis-à-vis angulation without biopsy instrument was: minimum 0.6° (retrograde, catheter) and maximum 8.5° (anterograde, needle).
Comparison of the biopsy instruments (VE OD 5.1 mm)
The extended needle and catheter were both associated with a clearly greater angulation in comparison with forceps, namely: anterograde 180.5° / 187.5° versus 167.0° and retrograde 102.3° / 100.8° versus 90.7°. In comparison with the flush instrument, extension of the forceps was associated with a decrease in anterograde angulation of 9.6°.
Comparison of the biopsy instruments (VE OD 3.7 mm)
In the extended state a clearly greater angulation, both anterograde and retrograde, was achieved with the forceps OD 0.8 mm in comparison with the forceps OD 1.0 mm, namely 157.3° / 90.6° versus 141.4° / 64.3°, respectively.
Comparison of the bronchoscopes
The largest anterograde angle with biopsy instrument in the working channel was measured with the VE OD 5.1 mm for needle and catheter (180.5° and 190.1°), and retrograde with the VE OD 6.2 mm, again for needle and catheter (108.3° and 110.1°). The anterograde angles measured for the VE OD 3.7 mm with forceps in the channel were at least 9.7° smaller than those measured for the VE OD 5.1 mm, and the retrograde angles at least 5.1° smaller than those measured for the VE OD 6.2 mm.
Wear of the bronchoscopes
The angulation of the bronchoscope without a biopsy instrument was less by a maximum of 23.1° than that specified by the manufacturer. The latter expect that major repairs will be needed after some 1,500 examinations. Although the 'scopes employed in our investigation had been used in the clinical setting on only some 500 occasions, this sufficed to reduce their angulation properties.
Advantage of extending the biopsy instrument
Maximum angulation was seen with needle and catheter
In comparison with forceps and cryoprobe, the largest increase in maximum angulation - up to 28° anterograde, and up to 11.6° retrograde - was measured with needle and catheter (Table 1).
Both the catheter and the plastic sheath enclosing and guiding the needle are less stiff than the metal instruments themselves. This can clearly be seen on further extending the biopsy instrument (catheter, forceps, needle) beyond the working channel of the bronchoscope (Figure 2a-c).
The largest angulation is seen with the 5.1 mm bronchoscope
Comparison of the bronchoscopes with biopsy instruments in the working channel revealed the largest anterograde angulation for the VE OD 5.1 mm - maximum 190.1° - and the largest retrograde angulation for the VE OD 6.2 mm - maximum 110.1° - again measured with needle or catheter in the channel (Table 1).
While the manufacturer's specifications with working channel empty indicate a larger anterograde angulation for the VE OD 5.1 mm (210°) than for the VE OD 6.2 mm (180°), the larger angulation also seen with needle or catheter in the channel is due to the fact that in the case of the VE OD 5.1 mm the same needle in a small-calibre sheath (OD 1.8 mm), and the small-calibre aspiration catheter (OD 1.8 mm) were used. The explanation of the fact that retrograde angulation with no biopsy instrument in the channel is equal (130°) for both VE OD 5.1 mm and VE OD 6.2 mm, is that the VE OD 6.2 mm 'scope is provided with a stronger bending control wire capable of exerting a greater traction against the greater resistance of the catheter with the larger outside diameter. For the same reason, angulation of the VE OD 3.7 mm (anterograde angulation 210°, retrograde 130°) with forceps in the working channel is smaller than that of the CE OD 5.1 mm (Table 1). For the former bronchoscope only mini-forceps are available.
In the diagnostic workup of peripheral pulmonary carcinoma, flexible bronchoscopy is of particular importance since surgery is often no longer an option, either because the tumour is too far advanced or the patient would not survive the operation [1, 12, 13]. Transbronchial biopsy also makes it possible to intubate acutely angled subsegmental ostia, for which purpose both anterograde and retrograde angulation is indispensible.
The largest angles with biopsy instrument in the working channel were measured for the catheter, and - somewhat smaller - for the needle. Since, however, these instruments serve a cytological purpose, the diagnostic usefulness of the material thus obtained - in particular with regard to a need to identify an EGFRmutation - must be considered low . On the other hand, the use of the cell block technique may be expected to improve the diagnosis - in particular in the case of the 2.3 mm catheter . However, the CE OD 5.1 mm 'scope used with the 1.8 mm catheter or needle provides excellent anterograde angulation and can thus also be used. The option of pre-bending the distal end of the catheter prior to TBCA may - in comparison with TBB and TBNA - also promote further advancement and correct positioning of the catheter in the peripheral lung (Figure 2a).
The appreciably better bending properties of catheter and needle in comparison with forceps, and possibly differences in the handling characteristics of the instruments in the periphery should be taken into account when preparing for a transbronchial biopsy.
The authors Karl-Josef Franke, Georg Nilius, Christiane Bruckner and Karl-Heinz Ruhle declare that no financial or other potential conflicts of interest exist in the subject matter.
- Chechani V: Bronchoscopic diagnosis of solitary pulmonary nodules and lung masses in the absence of endobronchial abnormality. Chest 1996,109(3):620–5. 10.1378/chest.109.3.620PubMedView ArticleGoogle Scholar
- Bilaçeroglu S, Kumcuoglu Z, Alper H, Osma E, Cagirici U, Günel O, Bayol U, Celikten E, Perim K, Köse T: CT bronchus sign-guided bronchoscopic multiple diagnostic procedures in carcinomatous solitary pulmonary nodules and masses. Respiration 1998,65(1):49–55. 10.1159/000029237PubMedView ArticleGoogle Scholar
- Franke KJ, Theegarten D, Hann von Weyhern C, Nilius G, Brueckner C, Hetzel J, Hetzel M, Ruhle KH, Enderle MD, Szyrach MN: Prospective controlled animal study on biopsy sampling with new flexible cryoprobes versus forceps: evaluation of biopsy size, histological quality and bleeding risk. Respiration 2010,80(2):127–32. 10.1159/000287251PubMedView ArticleGoogle Scholar
- Babiak A, Hetzel J, Krishna G, Fritz P, Moeller P, Balli T, Hetzel M: Transbronchial cryobiopsy: a new tool for lung biopsies. Respiration 2009,78(2):203–8. 10.1159/000203987PubMedView ArticleGoogle Scholar
- Katis K, Inglesos E, Zachariadis E, Palamidas P, Paraskevopoulos I, Sideris G, Tamvakopoulou E, Apostolopoulou F, Rasidakis A: The role of transbronchial needle aspiration in the diagnosis of peripheral lung masses or nodules. Eur Respir J 1995,8(6):963–6.PubMedGoogle Scholar
- Gasparini S, Ferretti M, Secchi EB, Baldelli S, Zuccatosta L, Gusella P: Integration of transbronchial and percutaneous approach in the diagnosis of peripheral pulmonary nodules or masses. Experience with 1,027 consecutive cases. Chest 1995,108(1):131–7. 10.1378/chest.108.1.131PubMedView ArticleGoogle Scholar
- Franke KJ, Nilius G, Ruhle KH: [Transbronchial biopsy in comparison with catheter aspiration in the diagnosis of peripheral pulmonary nodules]. Pneumologie 2006,60(1):7–10. 10.1055/s-2005-919107PubMedView ArticleGoogle Scholar
- Shure D: Transbronchial biopsy and needle aspiration. Chest 1989,95(5):1130–8. 10.1378/chest.95.5.1130PubMedView ArticleGoogle Scholar
- Franke KJ, Nilius G, Ruhle KH: Transbronchial catheter aspiration compared to forceps biopsy in the diagnosis of peripheral lung cancer. Eur J Med Res 2009,14(1):13–7.PubMed CentralPubMedView ArticleGoogle Scholar
- Yoshikawa M, Sukoh N, Yamazaki K, Kanazawa K, Fukumoto S, Harada M, Kikuchi E, Munakata M, Nishimura M, Isobe H: Diagnostic value of endobronchial ultrasonography with a guide sheath for peripheral pulmonary lesions without X-ray fluoroscopy. Chest 2007,131(6):1788–93. 10.1378/chest.06-2506PubMedView ArticleGoogle Scholar
- Eberhardt R, Anantham D, Ernst A, Feller-Kopman D, Herth F: Multimodality bronchoscopic diagnosis of peripheral lung lesions: a randomized controlled trial. Am J Respir Crit Care Med 2007,176(1):36–41. 10.1164/rccm.200612-1866OCPubMedView ArticleGoogle Scholar
- Goeckenjan G, Sitter H, Thomas M, Branscheid D, Flentje M, Griesinger F, Niederle N, Stuschke M, Blum T, Deppermann KM, Ficker JH, Freitag L, Lübbe AS, Reinhold T, Späth-Schwalbe E, Ukena D, Wickert M, Wolf M, Andreas S, Auberger T, Baum RP, Baysal B, Beuth J, Bickeböller H, Bödking A, Bohle RM, Brüske I, Burghuber O, Dickgreber N, Diederich S, Dienemann H, Eberhardt W, Eggeling S, Fink T, Fischer B, Franke M, Friedel G, Gauler T, Gütz S, Hautmann H, Hellmann A, Hellwig D, Herth F, Heussel CP, Hilbe W, Hoffmeyer F, Horneber M, Huber RM, Hübner J, Kauczor HU, Kirchbacher K, Kirsten D, Kraus T, Lang SM, Martens U, Mohn-Staudner A, Müller KM, Müller-Nordhorn J, Nowak D, Ochmann U, Passlick B, Petersen I, Pirker R, Pokrajac B, Reck M, Riha S, Rübe C, Schmittel A, Schönfeld N, Schütte W, Serke M, Stamatis G, Steingräber M, Steins M, Stoelben E, Swoboda L, Teschler H, Tessen HW, Weber M, Werner A, Wichmann HE, Irlinger Wimmer E, Witt C, Worth H: Prevention, diagnosis, therapy, and follow-up of lung cancer: interdisciplinary guideline of the German Respiratory Society and the German Cancer Society. Pneumologie 2011,65(1):39–59. 10.1055/s-0030-1255961PubMedView ArticleGoogle Scholar
- Chhajed PN, Tamm M: Bronchoscopy for small pulmonary nodules and mediastinal staging of lung cancer: just do it! Am J Respir Crit Care Med 2006,174(9):961–2. 10.1164/rccm.200607-925EDPubMedView ArticleGoogle Scholar
- Mok TS, Wu YL, Thongprasert S, Yang CH, Chu DT, Saijo N, Sunpaweravong P, Han B, Margono B, Ichinose Y, Nishiwaki Y, Ohe Y, Yang JJ, Chewaskulyong B, Jiang H, Duffield EL, Watkins CL, Armour AA, Fukuoka M: Gefitinib or carboplatin-paclitaxel in pulmonary adenocarcinoma. N Engl J Med 2009,361(10):947–57. 10.1056/NEJMoa0810699PubMedView ArticleGoogle Scholar
- Natu S, Hoffman J, Siddiqui M, Hobday C, Shrimankar J, Harrison R: The role of endobronchial ultrasound guided transbronchial needle aspiration cytology in the investigation of mediastinal lymphadenopathy and masses, the North Tees experience. J Clin Pathol 2010,63(5):445–51. 10.1136/jcp.2009.074328PubMedView ArticleGoogle Scholar